- Issue Time
Installation of bimetallic thermometer with every angle connection
Positioning the stem before installation, the stem should be set to the desired angle as follows:
Figure 1: Loosen the four screws labeled “A” and “B” in Figure 1, until the harness revolves freely without twisting the flexible housing.
Figure 2: While holding the case, revolve the harness clockwise or counterclockwise, as indicated by arrows in Figure 2, to place the harness in a position that will permit flexing the stem in the desired direction with respect to the case. Then lock the two screws labeled “A.”
Figure 3: Flex the stem to the desired angle with respect to the face of the thermometer, as shown in Figure 3, then lock the two screws labeled “B.”
To assure longest life, the “EVERY- ANGLE” joint should be operated only when necessary during installation or removal of the thermometer with every angle conn.